Complementary to the regional broadband network, a deployment of borehole-installed, wide-dynamic range seismographic stations is being established along the Hayward Fault and throughout the San Francisco Bay toll bridges network. This network is a cooperative development of the BSL and the USGS, with support from USGS, Caltrans, EPRI, the University of California Campus/Laboratory Collaboration (CLC) program, LLNL, and LBNL (Figure 4.1 and Table 4.1). Efforts at ongoing development of the network have also recently been enhanced by through coordinated efforts with the Mini-PBO project (Chapter 7, which is partially funded by NSF and by the member institutions of that project).
The purpose of the network is threefold: 1) to lower substantially the threshold of microearthquake detection, 2) to increase the recorded bandwidth for events along the Hayward fault, and 3) to obtain bedrock ground motion signals at the bridges from small earthquakes for investigating bridge responses to stronger ground motions. A lower detection threshold increases the resolution of the fault-zone seismic structure; allows seismologists to monitor the spatial and temporal evolution of seismicity at magnitudes down to , where earthquake rates are many times higher than those captured by surface sites; allows researchers to look for pathologies in seismicity patterns that may be indicative of the nucleation of large damaging earthquakes; and allows scientists to investigate fault and earthquake scaling, physics and processes in the San Francisco Bay Area. This new data collection will also contribute to improved working models for the Hayward fault. The bedrock ground motion recordings are also being used to provide input for estimating the likely responses of the bridges to large, potentially damaging earthquakes. Combined with the improved Hayward fault models, source-specific response calculations can be made, as well.
The Hayward Fault Network (HFN) consists of two parts. The Northern Hayward Fault Network (NHFN) is operated by the BSL and currently consists of 25 stations, including those located on Bay Area bridges and at borehole sites of the Mini-PBO (MPBO) project. This network is considered part of the BDSN and uses the network code BK. The Southern Hayward Fault Network (SHFN) is operated by the USGS and currently consists of 5 stations. This network is considered part of the NCSN and uses the network code NC. This chapter is primarily focused on the NHFN and activities associated with the BSL operations.
The five MPBO sites have 3-component borehole geophone packages. All the remaining HFN sites have six-component borehole sensor packages. The packages were designed and fabricated at LBNL's Geophysical Measurement Facility by Don Lippert and Ray Solbau, with the exception of site SFAB. For the HFN sites, three channels of acceleration are provided by Wilcoxon 731A piezoelectric accelerometers and three channels of velocity are provided by Oyo HS-1 4.5 Hz geophones. Velocity measurements for the MPBO sites are provided by Mark Products L-22 2 Hz geophones (Table 4.2). The 0.1-400 Hz Wilcoxon accelerometers have lower self-noise than the geophones above about 25-30 Hz, and remain on scale and linear to 0.5 g. In tests performed in the Byerly vault at UC Berkeley, the Wilcoxon is considerably quieter than the FBA-23 at all periods, and is almost as quiet as the STS-2 between 1 and 50 Hz.
Sensors are generally installed at depths of about 100 m, but several sites have sensors emplaced at depths of over 200 m and the Dumbarton bridge sites have sensors at multiple depths (Table 4.1). During initial stages of the project, the NHFN sensors provided signals to on-site Quanterra Q730 and RefTek 72A-07 data loggers.
Today, twelve of the NHFN sites currently have Quanterra data loggers with continuous telemetry to the BSL. Similar to BDSN sites, these stations are capable of on-site recording and local storage of all data for more than one day and have batteries to provide backup power. Signals from these stations are digitized at a variety of data rates up to 500 Hz at 24-bit resolution (Table 4.3). In contrast to the BDSN implementation, the NHFN data loggers employ casual FIR filters at high data rates and acausal FIR filters at lower data rates. Because of limitations in telemetry bandwidth and disk storage, 8 of these sites transmit triggered data at 500 sps, using the Murdock, Hutt, and Halbert (MHH) event detection algorithm (Murdock and Hutt, 1983), and continuous data at reduced rates (100, 20 and 1 sps) to the BSL, while the four MPBO sites transmit continuous data at 100 sps (one MPBO site does not have telemetry yet).
The remaining 12 sites of the NHFN have in the past recorded data using RefTek data loggers. These sites do not have continuous telemetry for acquisition and required visits from BSL staff for data recovery. Collection of data from these sites has been discontinued, but efforts are underway to upgraded them with Quanterra Q4120 data loggers and continuous telemetry.
Signals from the 5 SHFN stations are digitized by Nanometrics data loggers at 100 sps and transmit continuous data to Menlo Park by radio. These digital data streams are processed by the Earthworm system with the NCSN data and waveforms are saved when the Earthworm detects an event.
Experience has shown that the MHH detector does not provide uniform triggering across the NHFN on the smallest events of interest. In order to insure the recovery of 500 sps data for these earthquakes, a central-site controller has recently been implemented at the BSL using the 500 sps vertical component geophone data for event detection. Originally the 100 sps vertical component geophone data was used for event detection but the bandwidth proved to be inadequate for detection of the smaller events where most of the seismic wave energy was at frequencies above 40 Hz. Triggers from this controller are being used to recover the 500 sps data from the NHFN data loggers.
Data from the NHFN and SHFN are archived at the NCEDC. At this time, the tools are not in place to archive the Hayward fault data together. The NHFN data are archived with the BDSN data, while the SHFN are archived with the NCSN data (Chapter 10). However, the new central-site controller will provide the capability to both include SHFN data in the event detection and extract SHFN waveforms for these events in the future.
As originally planned, the Hayward Fault Network was to consist of 24 to 30 stations, 12-15 each north and south of San Leandro, managed respectively by UCB and USGS. This is not happening quickly, although west of the fault, Caltrans has provided sites along the Bay bridges. This important contribution to the Hayward Fault Network has doubled the number of sites with instrumentation. At times, Caltrans provides holes of opportunity away from the bridges (e.g., HERB), so we have plans for additional stations that will bring the network geometry to a more effective state for imaging and real-time monitoring of the fault.
As a check on the calibration and an example of the capabilities of a borehole installed network, we compare the bandpass filtered (0.4-4 Hz) ground velocity data recorded at NHFN and MPBO stations for a M 7.3 deep focus teleseism that occurred in the vicinity of the Fiji Islands at a depth of 582 km. in Figure 4.2.
On average the MPBO NHFN sites are more consistent and quieter (Figure 7.4). This is due in large part to the greater depth of the MPBO sensors, the locations of MPBO stations in regions of generally less industrial and other cultural noise sources, and possibly to the absence of powered sensors (i.e. accelerometers) in their borehole sensor packages.
One of the most pervasive problems at NHFN stations equipped with the Q4120 data loggers is power line noise (60 Hz and its harmonics at 120, 180, and 240 Hz). This noise reduces the sensitivity of the MHH detectors. Whenever a NHFN station is visited, the engineer at the site and a seismologist at the BSL work together to expedite the testing process, especially when attempting to identify and correct ground-loop faults which generally induce significant 60, 120, 180, and 240 Hz seismic signal contamination due to stray power line signal pickup, generally inductively coupled and aggravated by the presence of ground loops.
Below is a synopsis of maintenance efforts performed over the past year for several NHFN stations that gives some idea of the ongoing maintenance and performance enhancing measures that we are continuing to implement.
Generally speaking, the accelerometers, being an active device, are more accurate and also more stable that the geophones so it is reasonable to assume that the most likely reason for the difference is that the assumed generator constants for the geophones are not accurate. Rodgers et al. (1995) describe a way to absolutely calibrate the geophones in situ and to determine their generator constant, free period and fraction of critical damping. The only external parameter that is required is the value of the geophones inertial mass.
We have built a calibration test box which allows us to routinely perform the testing described by Rodgers et al. whenever site visits are made. The box drives the signal coil with a known current step and rapidly switches the signal coil between the current source and the data logger input. From this information, expected and actual sensor response characteristics can be compared and corrections applied. Also, changes in the sensor response over time can be evaluated so that adjustments can be made and pathologies arising in the sensors due to age can be identified. Once a geophone is absolutely calibrated, we can also check the response of the corresponding accelerometer.
We are now performing the initial calibration tests and response adjustments for all NHFN stations as sites are visited for routine maintenance. We also plan a scheduled re-tests of all sites to monitor for sensor responses changes through time.
The NHFN station hardware has proven to be quite reliable and half of the currently operating stations (BRIB, CMSB, CRQB and SMCB) did not require a site visit during the past year. The following stations were visited to perform various maintenance and/or repair operations.
BBEB: Prepped BBEB hardware for installation of station at pier E7. Data will be telemetered from pier E7 to BBEB via FreeWave 900 MHz radio and forwarded by the Q4120 to the BSL via 2.4 GHz radio along with the BBEB data. Ran Wilan radio tests on simultaneous links to BBEB and W02B.
HERB: Revamped the FRAD power supply and installed a twist-lock plug on the input power cable.
RFSB: Upgraded Q4120 data logger with installation of Q730PWR board and new software. Set battery charge voltage to 13.6 Vdc. Upgraded Q730 with software version is MS001122AL.bo, and config version is mscfg011004.lzh. Changed Q730 datalogger configuration of the B?? channels from 20 Hz to 40 Hz in support of USArray.
W02B: Installed grounding clamp and changed wiring to reduce 60 Hz noise on velocity channel. Extended DC cables of AC adapters for the FRAD and Cylink radio so that they will reside in the battery box instead of the electronics box. Replaced blown fuse on battery positive cable with a 7.5A. Realigned and securely clamped antenna at McCone roof end of the W02B telemetry link to stop it from moving in high winds. Changed the AC power adapter on the McCone master Cylink radio. Encountered problems with GPS clock not locking and subsequently fixed problem using Quanterra supplied procedure to reset the GPS clock.
Caltrans has also provided funding for instrumentation of several other land sites which we will install as future Caltrans drill time becomes available. Currently we are considering sites for these additional holes-of-opportunity at Pt. Pinole, on Wildcat Mtn. in the north Bay. We are in the process of obtaining permission from the East Bay Regional Park District (EBRPD) to site a HFN station at the Point Isabel Regional Shoreline.
Pattern Recognition: In order to improve the detection and analysis of small events (down to -1.0) some specialized algorithms are being developed. The Murdock-Hutt detection algorithms used by MultiSHEAR, which basically flags an event whenever the short-term average exceeds a longer-term average by some threshold ratio, is neither appropriate for nor capable of detecting the smallest seismic events. One solution is to use a pattern recognition approach to identify small events associated with the occurrence of an event which was flagged by the REDI system. Tests have indicated that the pattern recognition detection threshold is -1.0 for events occurring within 10 km of a NHFN station. The basic idea is to use a quarter second of the initial P-wave waveform, say, as a master pattern to search for similar patterns that occur within one day, say, of the master event. Experimentally, up to six small CMSB recorded events, at the -1.0 threshold and occurring within one day of a master pattern, have been identified.
The pattern recognition method is CPU intensive, however, and it will require a dedicated computer to handle the pattern recognition tasks. To expedite the auto-correlation processing of the master pattern, an integer arithmetic cross-correlation algorithm has been developed which speeds up the requisite processing by an order of magnitude.
Phase Onset Time Detection: The phase onset time detector makes use of the concept that the complex spectral phase data, over the bandwidth of interest (i.e., where the SNR is sufficiently high), will sum to a minimum at the onset of an impulsive P-wave. The algorithm searches for the minimum phase time via phase shifting in the complex frequency domain over the bandwidth where the SNR is above 30 dB, say, to identify the onset time of the seismic phase. The algorithm requires that the recorded waveforms be deconvolved to absolute ground displacement. This implicitly requires that any acausality in the anti-aliasing filtration chain, such as the FIR filters used in the BDSN Quanterra data loggers, be removed. The algorithm typically resolves P-wave onset times to one-fiftieth of the sample interval or better.
Phase Coherency: A spectral phase coherency algorithm was developed to facilitate high resolution quantification of the similarities and differences between highly similar Hayward fault events which occur months to years apart. The resolution of the complex spectral phase coherency methodology is an order of magnitude better that the cross correlation method which is commonly used to identify highly similar events with resolution of order a few meters. This method, originally developed using NHFN borehole data, is now being applied as well to data from another borehole network (the HRSN) to provide more rapid and objective identification of the large fraction (approx. 40) of characteristically repeating microearthquakes that occur at Parkfield, CA.
On October 19 and 20, 2003, two earthquakes with 3.5 and 3.4 occurred ENE of Orinda, CA. Fortuitously, their hypocenters were located almost directly below the northern HFN borehole station at Briones (BRIB). While the automatic system report depths of approximately 10 km for these two events, the S-P times at BRIB indicates that they must be located at depths of less than 5 km. More than 4000 fore- and aftershocks were recorded, during the first week of the sequence, with magnitudes down to -2. At the beginning of December, 2003 the aftershocks were continuing at a rate of six per day with a 2.9 occurring on January 1, 2004.
This sequence, which has events spanning six orders of magnitude, is ideal for tesing the pattern recognition, phase onset time detection and phase coherence algorithms discussed above.
Under Bob Nadeau's, Bob Uhrhammer's and Doug Dreger's general supervision, Rich Clymer, Wade Johnson, Doug Neuhauser, Bill Karavas, John Friday, and Dave Rapkin all contribute to the operation of the NHFN. Bob Nadeau, Bob Uhrhammer and Lind Gee contributed to the preparation of this chapter.
Partial support for the NHFN is provided by the USGS through the NEHRP external grant program. Expansion of the NHFN has been made possible through generous funding from Caltrans, with the assistance of Pat Hipley. Larry Hutchings of LLNL has been an important collaborator on the project.
Rogers, P.W., A.J. Martin, M.C. Robertson, M.M. Hsu, and D.B. Harris, Signal-Coil Calibration of Electromagnetic Seismometers, Bull. Seism. Soc. Am., 85(3), 845-850, 1995.
Murdock, J., and C. Hutt, A new event detector designed for the Seismic Research Observatories, USGS Open-File-Report 83-0785, 39 pp., 1983.
Berkeley Seismological Laboratory
215 McCone Hall, UC Berkeley, Berkeley, CA 94720-4760
Questions or comments? Send e-mail: email@example.com
© 2004, The Regents of the University of California