SUTB - Sutter Buttes, CA, USA
|Station Name||SUTB (SUTB)|
|In Operation||2005/10/16 00:00:00 - Present|
Site is located at Sutter Buttes in Sutter, CA.
This station was installed in May 2005 by USArray personnel as a collaboration with the Berkeley Seismological Laboratory, and was part of the USArray component of the Earthscope program. When the Travelling Array left California in 2007, the BSL adopted this station. It has been operating as a BDSN station since May 2008.
NetworkBK, BARD (Berkeley Digital Seismic Network and Bay Area Regional Deformation Network)
Vault/Borehole ConditionsThe SUTB installation is based on the USArray vault design.
Plots & Figures
View from SUTB seismic station
Typical USArray Installation
- Time series processing at the BSL
Last year Clean Clean & detrended Raw Raw & detrended Lifetime Clean Clean & detrended Raw Raw & detrended Text file (final results only) Clean Raw
GPS Station Pictures
Table of Seismic Instrumentation
Sensor Datalogger SEED Channels Location Broadband Streckeisen STS-2 VBB Tri-Axial Seismometer Q330HR BH?,HH?,LH?,VH? 00 Accelerometer Kinemetrics FBA ES-T Accel. (2 g max 10 v/g) Q330HR HN? 00 Receiver Antenna Radome - GPS Trimble NetRS Ashtech Choke-ring SCIGN Short - Sensor Datalogger SEED Channels Location Auxiliary AntennaCurrent Q330HR VE? 00 ClockPhase Q330HR LC?,VC? 00 ClockQuality Q330HR LC?,VC? 00 InputVoltage Q330HR VE? 00 SystemCurrent Q330HR VE? 00 SystemTemperature Q330HR VK? 00 VCOValue Q330HR VC? 00
Data & Metadata
Waveforms and associated metadata, and GPS data, are available at the Northern California Earthquake Data Center (NCEDC).
- BK Information Directory, including
Data completeness is defined both as "Completeness of observations" and "Cycle slips per observation". "Completeness of Obs." is the number of epochs in the final RINEX file normalized to the expected number. This percentage will go down if time is missing from the RINEX file. "Cycle slips per Obs." is the total number of detected cycle slip normalized to the total number of observations in the RINEX file. This number will increase as the receiver loses lock on satellites more frequently.
Previous Year Lifetime
The effects of multipath on the data are estimated by parameters for L1 and L2 (MP1 and MP2 respectively); see Estey and Meertens (GPS Solutions, 1999) for derivation. The daily value is the RMS of MP1 and MP2 throughout the day and for all satellites. Higher values indicate a greater prevalence and/or strength of multipathing, i.e. objects on the ground are providing multiple reflection pathways from the satellite to antenna.
Previous Year Lifetime